RKI Eagle 2 Replacement H2S Sensor ES-87RW-H2S
|Product Weight (lbs.)||No|
I'm one of your customers and I have on board the vessel RKI Instrument EAGLE Portable H2S Gas Detector.
I need to measure mentioned H2S gas. After measurement reading is 100pmm MORE. It's mean, more than 100ppm.
Is it possible by using RKI Instrument EAGLE Portable H2S Gas Detector, make exact readings. For example Range till 200ppm or 1000ppm, higher than 100ppm.
Is there any possibility to change Range scale for H2S gas in the settings or by ordering special range H2S sensor.
Thank you in advance.
The RKI Eagle instrument does have the capability to do 0-1000ppm of Hydrogen Sulfide (H2S), but there are parts involved. There isn't a specific sensor that is for the 0-1000ppm range. We can arrange to do an upgrade on your instrument if you would like to change the current range. Please email back with serial number and address info, and we’ll get started on a quote.